[ntp:questions] How exact are the PPS pulses of common GPSses?
gsricalde at gmail.com
Wed Feb 13 15:21:44 UTC 2013
I asked something similar on the time-nuts list as I have a cheap
navigation receiver that seems to have a 2 us offset compared to a
I just got a GPS board with a LEA-6T mdule, I'm using the time mark
feature to compare external PPS sources to its internal timebase.
Trimble Acutime Gold: ~80 ns early
Symmetricom 58534A: ~100 ns late
Comparing to a realization of UTC is another thing, see
On Wed, Feb 13, 2013 at 9:29 PM, Ralph Aichinger <ralph at pangea.at> wrote:
> Has anybody checked the PPS output for constant shift against
> UTC? Is it possible that the pulses are off with a constant offset,
> even if they have very little jitter? Or is this unlikely from what
> happens inside the receiver?
> /ralph -- I am waiting for the first tutorial how to build a
> rubidium-disciplined Raspberry Pi ;)
> questions mailing list
> questions at lists.ntp.org
More information about the questions